Standard Test Method for Measuring Optical Reflectivity of Transparent Materials
Automaticky preložený názov:
Štandardná skúšobná metóda pre meranie optickej odrazivosti priehľadné materiály
NORMA vydaná dňa 1.4.2016
Označenie normy: ASTM F1252-16
Dátum vydania normy: 1.4.2016
Kód tovaru: NS-636163
Počet strán: 4
Približná hmotnosť: 12 g (0.03 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
ICS Number Code 17.180.20 (Colours and measurement of light)
|Significance and Use|
4.1 Reflections from aircraft transparencies of instrument lights and other cockpit objects have been a concern to many pilots. Attempts to reduce these reflections have been hampered by the lack of a repeatable measurement method and variances in reflection measuring instrumentation.
4.2 This test method reduces the instrument variations by standardizing the light source, calculation method, and area of specimen surface being measured; a brand of instrumentation is not specified. Since the reflectivity is defined as the ratio of two luminance measurements and does not depend on an absolute measurement, dependence upon the accuracy of the calibration of the measuring instrument is reduced.
4.3 The test method may be used to objectively compare the reflection characteristics of various transparent materials. Furthermore, the test method may be used to evaluate reflections of a specified spectral distribution light source (for example, a monochromatic light-emitting diode) by using that source in place of the standard light source.
4.4 Provisions are made to check for polarization effects of the sample and to record the reflectivity of a standard specimen. These provisions are offered as an option to the tester; it is up to the user or the requiring agency to determine the significance and use of these data.
4.5 Since the reflections are measured photopically, the results are representative of what the pilot would visually perceive.
1.1 This test method covers a procedure for measuring the reflectivity of transparent materials, hereafter known as specimens. The results are repeatable without specifying a particular brand name of instrumentation.
1.2 This test method applies to substantially flat parts. Errors in measurement can occur if the parts being measured are not substantially flat.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Posledná aktualizácia: 2018-10-18 (Počet položiek: 2 186 691)
© Copyright 2018 NORMSERVIS s.r.o.