Standard Practice for Using Flame Photometric Detectors in Gas Chromatography
Štandardné praktiky pre použitie Flame fotometrických Detektory plynovou chromatografiou
NORMA vydaná dňa 1.1.2013
Označenie normy: ASTM E840-95(2013)
Dátum vydania normy: 1.1.2013
Počet strán: 12
Približná hmotnosť: 36 g (0.08 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
This practice is intended as a guide for the use of a flame photometric detector (FPD) as the detection component of a gas chromatographic system. The different principles of flame photometric detectors, and detector construction are presented in details. The detector sensitivity, minimum detectability, dynamic range, power law of sulphur response, linear range-phosphorus mode, unipower response range, noise and drift, and specificity are presented in details. The photomultiplier dark current is the magnitude of the FPD output signal measured with the FPD flame off. Flame background current is the difference in FPD output signal with the flame on and with the flame off in the absence of phosphorus or sulfur compounds in the flame.
flame ionization detector (FID), flame photometric detectors (FPD), gas chromatography (GC), packed columns, supercritical fluid chromatography (SFC), ICS Number Code 17.180.30 (Optical measuring instruments), 71.040.50 (Physicochemical methods of analysis)
1.1 This practice is intended as a guide for the use of a flame photometric detector (FPD) as the detection component of a gas chromatographic system.
1.2 This practice is directly applicable to an FPD that employs a hydrogen-air flame burner, an optical filter for selective spectral viewing of light emitted by the flame, and a photomultiplier tube for measuring the intensity of light emitted.
1.3 This practice describes the most frequent use of the FPD which is as an element-specific detector for compounds containing sulfur (S) or phosphorus (P) atoms. However, nomenclature described in this practice are also applicable to uses of the FPD other than sulfur or phosphorus specific detection.
1.4 This practice is intended to describe the operation and performance of the FPD itself independently of the chromatographic column. However, the performance of the detector is described in terms which the analyst can use to predict overall system performance when the detector is coupled to the column and other chromatographic system components.
1.5 For general gas chromatographic procedures, Practice E260 should be followed except where specific changes are recommended herein for use of an FPD.
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific safety information, see Section 4, Hazards.
|2. Referenced Documents|
Do detailu prepracovaný a praxou overený informačný systém s normalizačnou tématikou.
Mesačne aktualizované -> stále platné technické normy.
Chcete vedieť viac informácií ? Pozrite sa na túto stránku.
Posledná aktualizácia: 2018-08-18 (Počet položiek: 2 053 361)
© Copyright 2018 NORMSERVIS s.r.o.