Standard Test Method for Measuring Heat Flux Using Flush-Mounted Insert Temperature-Gradient Gages
NORMA vydaná dňa 1.9.2017
Označenie normy: ASTM E2683-17
Dátum vydania normy: 1.9.2017
Kód tovaru: NS-797373
Počet strán: 8
Približná hmotnosť: 24 g (0.05 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
heat-flux gage, temperature gradient, thermal transport,, ICS Number Code 17.200.10 (Heat. Calorimetry)
|Significance and Use|
5.1 The purpose of this test method is to measure the net heat flux to or from a surface location. For measurement of the radiant energy component the emissivity or absorptivity of the surface coating of the gage is required. When measuring the convective energy component the potential physical and thermal disruptions of the surface must be minimized and characterized. Requisite is to consider how the presence of the gage alters the surface heat flux. The desired quantity is usually the heat flux at the surface location without the presence of the gage.
5.1.1 Temperature limitations are determined by the gage material properties, the method of mounting the sensing element, and how the lead wires are attached. The range of heat flux that can be measured and the time response are limited by the gage design and construction details. Measurements of a fraction of 1 kW/m 2 to above 10 MW/m2 are easily obtained with current gages. With thin film sensors a time response of less than 10 μs is possible, while thicker sensors may have response times on the order of 1 s. It is important to choose the gage style and characteristics to match the range and time response of the required application.
5.1.2 When differential thermocouple sensors are operated as specified for one-dimensional heat flux and within the corresponding time response limitations, the voltage output is directly proportional to the heat flux. The sensitivity, however, may be a function of the gage temperature.
5.2 The measured heat flux is based on one-dimensional analysis with a uniform heat flux over the surface of the gage. Measurements of convective heat flux are particularly sensitive to disturbances of the temperature of the surface. Because the heat-transfer coefficient is also affected by any non-uniformities in the surface temperature, the effect of a small temperature change with location is further amplified as explained by Moffat et al. (FIG. 2 Diagram of an Installed Insert Heat-Flux Gage
5.2.1 If the gage is water cooled, the thermal pathway to the plate is less important. The heat transfer to the gage enters the water as the heat sink instead of the surrounding plate. Consequently, the thermal resistance between the gage and plate may even be increased to discourage heat transfer from the plate to the cooling water. Unfortunately, this may also increase the thermal mismatch between the gage and surrounding surface.
5.2.2 shows a heat flux gage mounted into a plate with the surface temperature of the gage of Ts and the surface temperature of the surrounding plate of Tp. As previously discussed, a difference in temperature between the gage and plate may also increase the local heat transfer coefficient over the gage. This amplifies the measurement error. Consequently, a well designed heat flux gage will keep the temperature difference between the gage surface and the plate to a minimum, particularly if any convection is being measured.
5.2.3 Under transient or unsteady heat transfer conditions a different thermal capacitance of the gage than the surrounding material may also cause a temperature difference that affects the measured heat flux. Independent measures of the substrate and the gage surface temperatures are advantageous for defining the heat transfer coefficient and ensuring that the gage thermal disruption is acceptably small.
5.3 The heat flux gages described here may also be water cooled to increase their survivability when introduced into high temperature environments. By limiting the rise in gage temperature, however, a large disruption of the measured heat flux may result, particularly if convection is present. For convection measurements to match the heat flux experienced by the surrounding surface, the gage temperature must match the temperature of that surface. This will usually require the surrounding surface to also be water cooled.
5.4 The time response of the heat flux sensor can be estimated analytically if the thermal properties of the thermal resistance layer are well known. The time required for 98 % response to a step input (where α is the thermal diffusivity of the TRL. Covering or encapsulation layers must also be included in the analysis. The calibrated gage sensitivity in applies only under steady-state conditions.
5.4.1 For thin-film sensors the TRL
material properties may be much different from those of bulk
materials. Therefore, a direct experimental verification of the
time response is desirable. If the gage is designed to absorb
radiation, a pulsed laser or optically switched Bragg cell can be
used to give rise times of less than 1 μs (). A rise time on the order of 5
μs can be provided in a convective flow with a shock tunnel
( ,5.4.2 Because the response of these gages
is close to an exponential rise, a measure of the first-order time
constant, τ, for the gage can be obtained by matching the
experimental response to step changes in heat flux with exponential
1.2 This general test method is quite broad in its field of application, size and construction. Two different gage types that are commercially available are described in detail in later sections as examples. A summary of common heat-flux gages is given by Diller (1.3 The values stated in SI units are to be regarded as the standard. The values stated in parentheses are provided for information only.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.5 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Chcete mať istotu, že používate len platné technické normy?
Ponúkame Vám riešenie, ktoré Vám zaistí mesačný prehľad o aktuálnosti noriem, ktoré používate.
Chcete vedieť viac informácií ? Pozrite sa na túto stránku.
Posledná aktualizácia: 2019-12-09 (Počet položiek: 2 346 957)
© Copyright 2019 NORMSERVIS s.r.o.