Standard Test Method for Determining Argon Concentration in Sealed Insulating Glass Units Using Spark Emission Spectroscopy
Automaticky preložený názov:
Štandardná skúšobná metóda pre stanovenie Argon koncentrácie v zapečatených izolačných skiel Použitie Spark emisnej spektroskopie
NORMA vydaná dňa 1.4.2012
Označenie normy: ASTM E2649-12
Dátum vydania normy: 1.4.2012
Kód tovaru: NS-45801
Počet strán: 6
Približná hmotnosť: 18 g (0.04 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
argon gas, fill gas, gas concentration, sealed insulating glass unit, spark emission spectroscopy (SES), ICS Number Code 71.040.50 (Physicochemical methods of analysis)
|Significance and Use|
This test method is intended to provide a means for determining the concentration of argon in sealed insulating glass units under controlled conditions in compliance with the apparatus manufacturer's instructions.
This is a non-destructive test method in that the edge seal of the test specimen is not breached in order to determine the argon gas concentration. However, damage to some glass coatings on the inner surfaces of the glass can occur.
This test method has been developed based on data collected in a controlled laboratory environment.
The device shall be used to determine the argon gas concentration in insulating glass units in a controlled laboratory environment. Refer to 12.3.
This test method may be used to determine the argon gas concentration before, during, or after the insulating glass unit is subjected to durability tests.
The accuracy of the test method is dependent upon the accuracy of the Spark Emission Spectroscope. When the concentration of the argon being measured is below certain levels, this test method is not applicable. See the spectroscope manufacturer’s literature for recommended levels of accuracy of a given model.
1.1 This test method covers procedures for using a spark emission spectroscope to determine the concentration of argon gas in the space between the lites of a sealed insulating glass unit.
1.2 This is a non-destructive test method.
1.3 This test method shall be used only in a controlled laboratory environment.
1.4 This test method is applicable for insulating glass units where argon has been added to the sealed insulating glass cavity and the balance of the gas is atmospheric air.
1.5 This test method is applicable for clear, double-glazed insulating glass units.
1.6 This test method is applicable for double-glazed insulating glass units with one lite having a metallic coating or tinted glass, or both, and with clear glass as the other lite.
1.7 This test method is applicable for triple-glazed insulating glass units only when the center lite of glass has a metallic coating (either low emissivity (low E) or reflective) and both of the other lites are clear glass.
1.8 This test method also includes a procedure for verifying the accuracy of the readings of the test apparatus.
1.9 The values stated in SI units are to be regarded as the standard. The values given in parentheses are mathematical conversions to inch-pound units that are provided for information only and are not considered standard.
1.10 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific warning statements, refer to 7.1.
|2. Referenced Documents|
Poskytovanie aktuálnych informácií o legislatívnych predpisoch vyhlásených v Zbierke zákonov od roku 1945.
Aktualizácia 2x v mesiaci !
Chcete vedieť viac informácii ? Pozrite sa na túto stránku.
Posledná aktualizácia: 2019-08-19 (Počet položiek: 2 262 021)
© Copyright 2019 NORMSERVIS s.r.o.