Standard Practice for Instrumented Indentation Testing
Automaticky preložený názov:
Štandardné praktiky pre Instrumented odsadenie testovanie
NORMA vydaná dňa 1.10.2015
Označenie normy: ASTM E2546-15
Dátum vydania normy: 1.10.2015
Kód tovaru: NS-622464
Počet strán: 24
Približná hmotnosť: 72 g (0.16 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
force displacement curve, indentation hardness, indentation modulus, indenter shape function, instrument compliance, instrumented indentation, zero point ,, ICS Number Code 19.100 (Non-destructive testing)
|Significance and Use|
5.1 IIT Instruments are used to quantitatively measure various mechanical properties of thin coatings and other volumes of material when other traditional methods of determining material properties cannot be used due to the size or condition of the sample. This practice will establish the basic requirements for those instruments. It is intended that IIT based test methods will be able to refer to this practice for the basic requirements for force and displacement accuracy, reproducibility, verification, reporting, etc., that are necessary for obtaining meaningful test results.
5.2 IIT is not restricted to specific test forces, displacement ranges, or indenter types. This practice covers the requirements for a wide range of nano, micro, and macro (see ISO 14577-1) indentation testing applications. The various IIT instruments are required to adhere to the requirements of the practice within their specific design ranges.
1.1 This practice defines the basic steps of Instrumented Indentation Testing (IIT) and establishes the requirements, accuracies, and capabilities needed by an instrument to successfully perform the test and produce the data that can be used for the determination of indentation hardness and other material characteristics. IIT is a mechanical test that measures the response of a material to the imposed stress and strain of a shaped indenter by forcing the indenter into a material and monitoring the force on, and displacement of, the indenter as a function of time during the full loading-unloading test cycle.
1.2 The operational features of an IIT instrument, as well as requirements for Instrument Verification ( ), Standardized Reference Blocks ( ) and Indenter Requirements ( ) are defined. This practice is not intended to be a complete purchase specification for an IIT instrument.
1.3 With the exception of the non-mandatory , this practice does not define the analysis necessary to determine material properties. That analysis is left for other test methods. includes some basic analysis techniques to allow for the indirect performance verification of an IIT instrument by using test blocks.
1.4 Zero point determination, instrument compliance determination and the indirect determination of an indenter’s area function are important parts of the IIT process. The practice defines the requirements for these items and includes non-mandatory appendixes to help the user define them.
1.5 The use of deliberate lateral displacements is not included in this practice (that is, scratch testing).
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|2. Referenced Documents|
Chcete mať istotu o platnosti využívaných predpisov?
Ponúkame Vám riešenie, aby ste mohli používať stále platné (aktuálne) legislatívne predpisy
Chcete vedieť viac informácií ? Pozrite sa na túto stránku.
Posledná aktualizácia: 2019-08-18 (Počet položiek: 2 261 931)
© Copyright 2019 NORMSERVIS s.r.o.