Standard Practice for Process Compensated Resonance Testing Via Swept Sine Input for Metallic and Non-Metallic Parts
Automaticky preložený názov:
Bežná prax pre procesné kompenzovaný rezonancie Testovanie Via sínusovým Vstup pre kovových a nekovových častí
NORMA vydaná dňa 1.12.2015
Označenie normy: ASTM E2534-15
Dátum vydania normy: 1.12.2015
Kód tovaru: NS-623612
Počet strán: 8
Približná hmotnosť: 24 g (0.05 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
damage identification, elastic properties, feature extraction, nondestructive examination, nondestructive inspection, process compensated resonant examination, process compensated resonant testing, production variation, quality control, resonance inspection, resonances, resonant frequency, resonant mode, resonant ultrasound spectroscopy, system health monitoring, vibration characteristics,, ICS Number Code 17.160 (Vibrations, shock and vibration measurement)
|Significance and Use|
5.1 PCRT Applications and Capabilities—PCRT has been applied successfully to a wide range of NDT applications in the manufacture and maintenance of metallic and non-metallic parts. Examples of anomalies detected are discussed in . PCRT has been shown to provide cost effective and accurate NDT solutions in many industries including automotive, aerospace, and power generation. Examples of successful applications currently employed in commercial use include, but are not limited to:
(1) Silicon nitride bearing elements
(2) Steel, iron, and aluminum rocker and control arms
(3) Gas turbine engine components (blades, vanes, disks)
(4) Cast cylinder heads and cylinder blocks
(5) Sintered powder metal gears and clutch plates
(6) Machined forged steel steering and transmission components (gears, shafts, racks)
(7) Ceramic oxygen sensors
(8) Silicon wafers
(9) Gears with induction hardened teeth
(10) Ceramic matrix composite (CMC) material samples and components
(11) Components with shot peened surfaces
(12) Machined or rolled-formed, or both, steel fasteners
(13) Additive manufactured components
5.2 General Approach and Equipment Requirements for PCRT via Swept Sine Input:
5.2.1 PCRT systems are comprised of hardware and software capable of inducing vibrations, recording the component response to the induced vibrations, and executing analysis of the data collected. Inputting a swept sine wave into the part has proven to be an effective means of introducing mechanical vibration, and can be achieved with a high quality signal generator coupled with an appropriate active transducer in physical contact with the part. Collection of the part’s frequency response can be achieved by recording the signal generated by an appropriate passive vibration transducer. The software required to analyze the available data may include a variety of suitable statistical analysis and pattern recognition tools. Measurement accuracy and repeatability are extremely important to the application of PCRT.
5.2.2 Hardware Requirements—A swept sine wave signal generator and response measurement system operating over the desired frequency range of the test part are required with accuracy better than 0.002 %. The signal generator should be calibrated to applicable industry standards. Transducers must be operable over same frequency range. Three transducers are typically used; one 5.3 Constraints and Limitations:
5.3.1 PCRT cannot separate parts based on visually detectable anomalies that do not affect the structural integrity of the part. It may be necessary to provide additional visual inspection of parts to identify these indications.
5.3.2 Excessive process variation of parts may limit the sensitivity of PCRT. For example, mass/dimensional variations exceeding 5 % may cause PCRT to be unusable.
5.3.3 Specific anomaly identification is highly unlikely. PCRT is a whole body measurement, and differentiating between a crack and a void in the same location is generally not possible. It may be possible to differentiate some anomalies by using multiple patterns and training sets.
5.3.4 PCRT will only work with stiff objects that provide resonances whose frequency divided by their width at half of the maximum amplitude (Q) are greater than 400 to 500. Although steel parts may be very stiff and perfectly reasonable to use for PCRT, steel foil would generally not be.
5.3.5 While PCRT can be applied to painted and coated parts in many cases, the presence of some surface coatings such as vibration-absorbing materials and heavy oil layers may limit or preclude the application of PCRT.
5.3.6 While PCRT can be applied to parts over a wide range of temperatures, it can not be applied to parts that are rapidly changing temperature. The part temperature should be stabilized before collecting resonance data.
5.3.7 Misclassified parts in the teaching set, along with the presence of unknown anomalies in the teaching set, can significantly reduce the accuracy and sensitivity of PCRT.
1.1 This practice describes a general procedure for using the process compensated resonance testing (PCRT) via swept sine input method to detect resonance pattern differences in metallic and non-metallic parts. The resonance differences can be used to distinguish acceptable parts with normal process variation from parts with material states and defects that will cause performance deficiencies. These material states and defects include, but are not limited to, cracks, voids, porosity, shrink, inclusions, discontinuities, grain and crystalline structure differences, density-related anomalies, heat treatment variations, material elastic property differences, residual stress, and dimensional variations. This practice is intended for use with instruments capable of exciting, measuring, recording, and analyzing multiple whole body, mechanical vibration resonance frequencies in acoustic or ultrasonic frequency ranges, or both.
1.2 This practice uses inch pound units as primary units. SI units are included in parentheses for reference only, and are mathematical conversions of the primary units.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|2. Referenced Documents|
Chcete mať istotu, že používate len platné technické normy?
Ponúkame Vám riešenie, ktoré Vám zaistí mesačný prehľad o aktuálnosti noriem, ktoré používate.
Chcete vedieť viac informácií ? Pozrite sa na túto stránku.
Posledná aktualizácia: 2019-08-18 (Počet položiek: 2 261 931)
© Copyright 2019 NORMSERVIS s.r.o.