Potrebujeme váš súhlas na využitie jednotlivých dát, aby sa vám okrem iného mohli ukazovať informácie týkajúce sa vašich záujmov. Súhlas udelíte kliknutím na tlačidlo „OK“.
Standard Test Method for Rockwell and Brinell Hardness of Metallic Materials by Portable Hardness Testers
Automaticky preložený názov:
Štandardná skúšobná metóda pre Rockwell a Brinell tvrdosť kovových materiálov od Frost & Sullivan
NORMA vydaná dňa 1.7.2014
Označenie normy: ASTM E110-14
Dátum vydania normy: 1.7.2014
Kód tovaru: NS-40436
Počet strán: 5
Približná hmotnosť: 15 g (0.03 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
This test method establishes the standard procedures, including the calibration, precision and bias of the apparatus used, for the determination of indentation hardness of metallic materials by means of portable hardness testers.
Brinell, hardness, portable, Rockwell, ICS Number Code 77.040.10 (Mechanical testing of metals)
|Significance and Use|
3.1 Portable hardness testers are used for testing materials that because of their size, location or other requirements such as test point are unable to be tested using traditional fixed instruments.
3.2 Portable hardness testers, by their nature, induce variation that could influence the test results; therefore, hardness measurements made in accordance with this test method are not considered to meet the requirements of or . The user should compare the results of the precision and bias studies in , and to understand the differences in results expected between portable and fixed instruments.
3.3 Two test parameters that can significantly influence the measurement accuracy when using portable hardness testers are the alignment of the indenter to the test surface and the timing of the test forces. The user is cautioned to do everything possible to keep the centerline of the indenter perpendicular to the test surface and to apply the test forces using the same time cycle as defined in Test Method or Test Methods .
3.4 Portable hardness testers are delicate instruments that are subject to damage when they are moved from one test site to another. Therefore, repeating the daily verification process during the testing sequence is recommended to insure that they are working properly.
3.5 Hardness testing at a specific location on a part may not represent the physical characteristics of the whole part or end product.
1.1 This test method defines the requirements for portable instruments that are intended to be used to measure the Rockwell or Brinell hardness of metallic materials by performing indentation tests on the surface of materials in the field or outside of a test lab, or in cases where the size or weight of the test piece prevents it from being tested on a standard or hardness tester.
1.2 The principles used to measure the Rockwell or Brinell hardness are the same as those defined in the standard test method for Rockwell or standard test method for Brinell.
Note 1: Standard test methods and will be referred to in this test method as the standard methods.
1.3 The portable hardness testers covered by this test method are verified only by the indirect verification method. Although the portable hardness testers are designed to employ the same test conditions as those defined in the standard test methods, the forces applied by the portable Rockwell and Brinell testers and the depth measuring systems of the portable Rockwell testers may not meet the tolerance requirements of the standard methods. Portable hardness testers shall use indenters that meet the requirements of the standard test methods.
1.4 This test method does not apply to portable hardness testers that measure hardness by a means or procedure that is different than those defined in or For example, this test method does not apply to the methods defined in ASTM standard Practice , Test Methods and or .
1.5 A report section is included to define how to indicate that the test result was obtained by using a portable device that conforms to this document.
1.6 is included that defines the periodic indirect verification and daily verification requirements for these instruments.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|2. Referenced Documents|
Posledná aktualizácia: 2022-06-30 (Počet položiek: 2 574 073)
© Copyright 2022 NORMSERVIS s.r.o.