Potrebujeme váš súhlas na využitie jednotlivých dát, aby sa vám okrem iného mohli ukazovať informácie týkajúce sa vašich záujmov. Súhlas udelíte kliknutím na tlačidlo „OK“.
Standard Test Method for Automated Measurement of Maturity, Fineness, Ribbon Width, and Micronaire of Cotton Fibers
NORMA vydaná dňa 1.7.2021
Označenie normy: ASTM D8394-21
Dátum vydania normy: 1.7.2021
Kód tovaru: NS-1035622
Počet strán: 7
Približná hmotnosť: 21 g (0.05 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
cotton, Cottonscope, fineness, gravimetric fineness, linear density, maturity, micronaire, ribbon width,
|Significance and Use|
5.1Cotton fiber that is on average finer and more mature is more desirable than coarse or immature fiber, although distinction between these qualities cannot be made quickly and accurately using current test methods. Immature fibers break more easily during processing and have a tendency to form into neps (small entanglements) during processing in the spinning mill. These consequences adversely affect yarn and fabric quality and appearance. Yarn and fabric produced from immature fiber is typically also less lustrous and does not take up dye consistently so yarn and fabric appearance may be different after dyeing.
5.2Maturity has a high positive correlation with fiber length and strength but genetic differences and differences in fiber wall thickness caused by plant diseases, soil, and water conditions during the growing season interfere with this relationship.
5.2.1Fine fibers are required for fine count yarn manufacture and fiber fineness affects yarn count, evenness and strength. Both fineness and ribbon width are strong genetic traits evident between species and affected by growing conditions within species.
5.2.2Micronaire has traditionally been used as a measure of fiber fineness although the value actually measures fiber specific surface area or surface area per unit weight. As a result, micronaire varies concomitantly with both maturity and fineness (see ). Maturity and fineness are related to micronaire via Lords equation.FIG. 2Relationship Between Micronaire (X), Fineness (H), and Maturity (M)
5.3Cottonscope values have been judged against fineness, maturity and ribbon width values produced by examination of thousands of individual, magnified fiber cross-sections. Relationships with equivalent values by these and other older test methods are highly significant5.4The precision of Cottonscope has been measured. Measurement of consistent quality, machine picked cotton using a test regime of one in three bales (with two specimens tested per bale sample), provides a precision of between 6.8 and 7.5 % for fineness, between 1.2 and 1.5 % for maturity, between 4 and 5% for micronaire and less than 1 % for ribbon width. These values incorporate sample and instrument variance but not inter-laboratory variance. Precision is improved by increasing the number of sub-samples measured per bale. The values for maturity and micronaire are similar to reported high volume instrument values.
5.4.1The method has not yet been widely controlled for acceptance testing. Inter-laboratory trials have shown agreement and precision can be good if instrument calibrations and test procedures are properly coincided. Calibration material should be conditioned before testing as directed in Practice .
5.4.2Cottonscope algorithms are calibrated using known cotton and fiber standards, for example, USDA AMS prepared calibration cottons, measured on a manufacturers standard instrument with values and the calibration material passed to the next instrument. It is advised if there are differences of practical significance between reported test values for two or more laboratories, comparative tests should be performed to determine any statistical bias between them, using competent statistical analysis. Ideally, these tests are performed using the same homogenous material.
1.1This test method covers the determination of linear density (gravimetric fineness hereafter stated as fineness), maturity, micronaire, and ribbon width of cotton fibers from a loose, chemically untreated sample taken before harvest, during ginning, during mill processing or unraveled from raw (undyed) yarn or fabric.
1.2This test method requires the use of an integrated instrument, for example, the Cottonscope (see ), that measures the maturity, fineness, ribbon width, and micronaire of cotton fiber.
FIG. 1Cottonscope Instrument
Note 1:For another test method to determine the maturity of cotton fibers, refer to Test Method .
1.3The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.5This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
Chcete mať istotu, že používate len platné technické normy?
Ponúkame Vám riešenie, ktoré Vám zaistí mesačný prehľad o aktuálnosti noriem, ktoré používate.
Chcete vedieť viac informácií ? Pozrite sa na túto stránku.
Posledná aktualizácia: 2022-06-30 (Počet položiek: 2 574 073)
© Copyright 2022 NORMSERVIS s.r.o.