Potrebujeme váš súhlas na využitie jednotlivých dát, aby sa vám okrem iného mohli ukazovať informácie týkajúce sa vašich záujmov. Súhlas udelíte kliknutím na tlačidlo „OK“.
Standard Test Method for Vanadium and Nickel in Crude and Residual Oil by X-ray Spectrometry (Includes all amendments and changes 10/1/2019).
NORMA vydaná dňa 1.8.2019
Označenie normy: ASTM D8252-19e1
Dátum vydania normy: 1.8.2019
Kód tovaru: NS-972060
Počet strán: 6
Približná hmotnosť: 18 g (0.04 libier)
Krajina: Americká technická norma
Kategória: Technické normy ASTM
crude, EDXRF, energy dispersive, energy dispersive X-ray fluorescence, MEDXRF, metals, monochromatic energy dispersive X-ray fluorescence, monochromatic wavelength dispersive X-ray fluorescence, MWDXRF, nickel, resid, residual oil, spectrometry, vanadium, wavelength dispersive, wavelength dispersive X-ray fluorescence, WDXRF, X-ray, X-ray fluorescence, XRF,
|Significance and Use|
5.1 This test method provides a rapid and precise elemental measurement with simple sample preparation. Typical analysis times are approximately 4 min to 5 min per sample with a preparation time of approximately 1 min to 3 min per sample.
5.2 The quality of crude oil is related to the amount of sulfur present. Knowledge of the vanadium and nickel concentration is necessary for processing purposes as well as contractual agreements.
5.3 The presence of vanadium and nickel presents significant risks for contamination of the cracking catalysts in the refining process.
5.4 This test method provides a means of determining whether the vanadium and nickel content of crude meets the operational limits of the refinery and whether the metal content will have a deleterious effect on the refining process or when used as a fuel.
1.1 This test method covers the quantitative determination of total vanadium and nickel in crude and residual oil in the concentration ranges shown in using X-ray fluorescence (XRF) spectrometry.
1.2 Sulfur is measured for analytical purposes only for the compensation of X-ray absorption matrix effects affecting the vanadium and nickel X-rays. For measurement of sulfur by standard test method use Test Methods , or other suitable standard test method for sulfur in crude and residual oils.
1.3 This test method is limited to the use of X-ray fluorescence (XRF) spectrometers employing an X-ray tube for excitation in conjunction with wavelength dispersive detection system or energy dispersive high resolution semiconductor detector with the ability to separate signals of adjacent and near-adjacent elements.
1.4 This test method uses inter-element correction factors calculated from XRF theory, the fundamental parameters (FP) approach, or best fit regression.
1.5 Samples containing higher concentrations than shown in must be diluted to bring the elemental concentration of the diluted material within the scope of this test method.
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6.1 The preferred concentrations units are mg/kg for vanadium and nickel.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.8 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
Posledná aktualizácia: 2022-07-01 (Počet položiek: 2 574 846)
© Copyright 2022 NORMSERVIS s.r.o.